Level gauge wholesale with kaidi86.com? The radar level gauge works by electromagnetic waves. Its working principle is to measure the specific liquid level by transmitting electromagnetic waves to the measured target. After the electromagnetic waves are emitted, they are reflected by the medium. For the radar level gauge, its key function is to ensure that it can transmit magnetrol guided wave radar signals smoothly. In our industrial production site, interference often occurs, so which interference sources will affect the measurement of the radar level gauge? let’s see. Discover more information at level switch manufacturer. Our Radar Level Meter has a range of up to 150 meters, frequency up to 120GHz and an accuracy of ±1mm, which can cope with various complex measurement conditions.
When the distance between the liquid level of the measured medium and the electromagnetic wave transmitter is less than 4m, the guided wave radar is selected. If the distance between the liquid level of the measured medium and the electromagnetic wave transmitter is greater than 4m and less than 35m, select the guided wave radar with the guided wave cable. When the distance between the liquid level of the measuring medium and the electromagnetic wave transmitter is greater than 35m, or when the liquid level of the measuring medium is high temperature, high viscosity or solid liquid level, the air-shooting radar should be selected.
If the radar level meter is unreasonably selected, the interference echo cannot be handled well, and the reliability of the instrument will be reduced. Therefore, the following factors should be considered when selecting a radar level meter: Conductivity and dielectric constant of the measured medium. The measured medium is a conductive liquid or a liquid with a dielectric constant above 4. Generally, a common radar is selected. Liquids with small dielectric constants (dielectric constants below 2) and some conductive solids often use precision radars or guided wave radars due to the large amount of interference echoes.
The influence of dust accumulation on the transmitter head of the radar level meter, due to the large amount of dust in the working condition of the radar level meter, the dust is easy to adhere, and affected by the ambient temperature, the dust is wet and easy to agglomerate, and the dielectric constant will become larger and adhered The transmitter head and the wall of the bell mouth of the radar level meter affect the accuracy and stability of the radar measurement.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Find even more info on https://www.kaidi86.com/.